with a diameter measured outer wall to outer wall of no greater than 42 mm and no less than 15 mm.
There have been five reports of graft thrombosis/occlusion during  global commercial use, each following treatment for BTAI.  One case resulted in patient death, and three cases resulted in  reintervention.
CYDECTIN (moxidectin) Injectable Solution for Beef and Nonlactating Dairy Cattle, Antiparasitic. Sterile.  Contains 10 mg moxidectin/mL (5,000 mg/500 mL).  Contents 500 mL.  Manufactured for:  Boehringer Ingelheim Vetmedica, Inc., St. Joseph, MO 64506 USA
Lack of sterility assurance:  Due to potential for bottles to leak.
Lot AH01VL1, Product Code 81449783, Expiry Date 31/12/2018
Nationwide Distribution - US including CA and VA.
PALLAS M/MAXIMIS Poly Screw 5.5 x 30mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Devices are not consistent with Quality System Requirements.
Catalog No. P355-030; Known Lot No. WNV120322
PALLAS M/MAXIMIS 40mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-040; Known Lot No. WNV101022.
PALLAS M/MAXIMIS Poly Screw Driver. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0010; Known Lot No. WNV111116.
PALLAS M/MAXIMIS Poly Screw 5.5 x 35mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P755-035; Known Lot No. WNV120322 WNV120413.
PALLAS M/MAXIMIS Cross Link Torque Handle. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0030; Known Lot No. WNV111109.
PALLAS M/MAXIMIS 70mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. M058-070; Known Lot No. WNV120322.
PALLAS M/MAXIMIS MIS Pedicle Cut. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Rod Pusher B. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Compressor. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0016; Known Lot No. WNV100630.
PALLAS M/MAXIMIS 7.5 x 45mm screw and 7.5 x 4.5 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P775-045; Known Lot No. Unknown, WNV120328, WNV120430, WNV110513.
PALLAS M/MAXIMIS 7.5 x 50mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM775-050; Known Lot No. Unknown, WNV120328, WNV120322.
PALLAS M/MAXIMIS 80mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-080; Known Lot No. WNV120322.
PALLAS M/MAXIMIS Anti Torque. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0014; Known Lot No. WNV110216.
PALLAS M/MAXIMIS Rod Checker and T-Bar. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Nut Driver 4.0 Hexa and Alignment Driver. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 6.5 x 40mm and 6.5 x 4.0 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM765-040; Known Lot No. Unknown, NHS150813, WNV120322.
PALLAS M/MAXIMIS Poly Screw 5.5 x  45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS355-045; Known Lot No. NHS150813, WNV101201.
PALLAS M/MAXIMIS Small Crosslink. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. CL47-000; Known Lot No. NHS160219.
PALLAS M/MAXIMIS 45mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-045; Known Lot No. WNV101015.
PALLAS M/MAXIMIS Poly Screw 5.5 x  40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS355-040; Known Lot No. NHS150813, WNV101201.
PALLAS M/MAXIMIS Poly Screw 7.5 x 45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P375-045; Known Lot No. WNV120328.
PALLAS M/MAXIMIS Reduction A. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0027; Known Lot No. WNV111118.
PALLAS M/MAXIMIS Inside Bender (L) and (R). Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0021; Known Lot No. WNV111130.
PALLAS M/MAXIMIS MISS/MIS Compressor and Rod Tightener. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Extension Guide Bush. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 250mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-250; Known Lot No. WNV111123.
PALLAS M/MAXIMIS 350 Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. SR60-350; Known Lot No. NHS160219.
PALLAS M/MAXIMIS Ratchet T Handle. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 35mm MIS rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. M058-035; Known Lot No. WNV110428 & Unknown.
PALLAS M/MAXIMIS MIS Screw 6.5 x 35mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM765-035; Known Lot No. Unknown.
PALLAS M/MAXIMIS Poly Screw 7.5 x 40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P375-040; Known Lot No. WNV120328 & WNV110525.
PALLAS M/MAXIMIS 40mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. MR58-040; Known Lot No. NHS150817.
PALLAS M/MAXIMIS 60mm Curved and Precontoured Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-060; Known Lot No. WNV120323, WNV120420.
PALLAS M/MAXIMIS Body Alignment Driver. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 50mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-050; Known Lot No. WNV100426.
PALLAS M/MAXIMIS 60mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-060; Known Lot No. WNV101109, WNV120322.
PALLAS M/MAXIMIS Head Holder. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 80mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-080; Known Lot No. WNV100210.
PALLAS M/MAXIMIS Guide Wire. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Extension Driver. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Poly Screw 6.5 x  45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS365-045; Known Lot No. NHS150813, WNV101206.
PALLAS M/MAXIMIS Poly Screw 7.5 x 55mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P375-055; Known Lot No. WNV120322.
PALLAS M/MAXIMIS S. Probe. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Crosslink 50-70mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 6.5 x 55mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P765-055; Known Lot No. WNV120322.
PALLAS M/MAXIMIS Poly Screw 5.5 x  55mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS355-055; Known Lot No. NHS160219.
PALLAS M/MAXIMIS Distractor. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0017; Known Lot No. WNV100630.
PALLAS M/MAXIMIS Poly Screw 6.5 x 45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P365-045; Known Lot No. WNV110513.
PALLAS M/MAXIMIS 5.5 mm Instrument Tap. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 6.5 x 50mm and 6.5 x 5.0 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM765-050; Known Lot No. Unknown, NHS160219, WNV120322.
PALLAS M/MAXIMIS 80mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. M058-080; Known Lot No. WNV111019 & WNV120424.
PALLAS M/MAXIMIS Set Screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 35mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-035; Known Lot No. NHS100106.
PALLAS M/MAXIMIS Poly Screw 5.5 x 50mm and 5.5 x 5.0 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P755-050; Known Lot No. WNV110513 and Unknown
PALLAS M/MAXIMIS 55mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. MR58-055; Known Lot No. NHS160219.
PALLAS M/MAXIMIS Tapper 6.5. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0005; Known Lot No. WNV111130.
PALLAS M/MAXIMIS 35mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. MR58-035; Known Lot No. NHS150611 & WNV120323.
PALLAS M/MAXIMIS 50mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. C060-050; Known Lot No. WNV110516
PALLAS M/MAXIMIS MIS Screw5.5 x 55mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM755-055; Known Lot No. NHS160219.
PALLAS M/MAXIMIS 35mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-035 and R060-040; Known Lot No. WNV120322.
PALLAS M/MAXIMIS Poly Screw 6.5 x 40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P355-040; Known Lot No. WNV110511.
PALLAS M/MAXIMIS 30 mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-030; Known Lot No. WNV120322.
PALLAS M/MAXIMIS Extension Block. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Closed Extension. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Marking Pin (Smooth). Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Compass. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS355-035; Known Lot No. WNV101201.
PALLAS M/MAXIMIS Rod Bender. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0015; Known Lot No. WNV111105.
PALLAS M/MAXIMIS MIS Tapper 5.5mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 45mm MIS . Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. M058-045; Known Lot No. WNV120322.
PALLAS M/MAXIMIS 70mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. S060-070; Known Lot No. WNV101110.
PALLAS M/MAXIMIS MIS Poly Screw Driver B. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. Unknown; Known Lot No. NHS160315.
PALLAS M/MAXIMIS Tapper 5.5. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0004; Known Lot No. WNV111130.
PALLAS M/MAXIMIS Dilator Handle. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0902-1002; Known Lot No. WNV120802.
PALLAS M/MAXIMIS 60mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. MR58-060; Known Lot No. NHS150611 & NHS160219.
Catalog No. M058-060; Known Lot No. WNV120322 & Unknown.
PALLAS M/MAXIMIS MIS Screw 6.5 x 55mm and 6.5 x 5.5 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM765-055; Known Lot No. Unknown, NHS160219.
PALLAS M/MAXIMIS 45mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. MR58-045; Known Lot No. WNV110128.
Catalog No. 0902-1025. Known Lot No. WNV120802.
PALLAS M/MAXIMIS Nut Inserter B. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Instrument, Known Lot No. WNV110318, WNV120809.
PALLAS M/MAXIMIS 35mm Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. SR60-035; Known Lot No. NHS100106.
PALLAS M/MAXIMIS Large Crosslink. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. CL60-100; Known Lot No. NHS160219.
PALLAS M/MAXIMIS Poly Screw 7.5 x 50mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P375-050; Known Lot No. WNV120322.
PALLAS M/MAXIMIS Open Extension (Large). Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 5.5 x 45mm & 5.5 x 4.0mm, and 7.5 x 45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Tapper 6.5mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS I Handle. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 5.5 x 40mm and 5.5 x 4.0 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM755-040; Known Lot No. NHS150813, WNV110513, WNV12064.
PALLAS M/MAXIMIS Poly Screw 6.5 x 50mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P365-050; Known Lot No. WNV110513.
PALLAS M/MAXIMIS Marking Pin (Round Jaw). Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 5.5 x 35mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 7.5 x 35mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P775-035; Known Lot No. WNV110525.
PALLAS M/MAXIMIS Poly Screw 6.5 x  50mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 400mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. SR60-400; Known Lot No. NHS160219.
PALLAS M/MAXIMIS Wall Check. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0007; Known Lot No. WNV111115.
PALLAS M/MAXIMIS 70mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-070; Known Lot No. WNV120322.
Catalog No. P755-030; Known Lot No. WNV120322.
PALLAS M/MAXIMIS 40mm MIS rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. M058-040; Known Lot No. WNV120322 & Unknown.
PALLAS M/MAXIMIS MISS/MIS Tapper 7.5 mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Open Extension (Small). Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Awl. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0703-0001; Known Lot No. WNV101126.
PALLAS M/MAXIMIS Polyaxial and MIS Screw, 5.5 x 45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 7.5 x 40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P775-040; Known Lot No. WNV110513.
PALLAS M/MAXIMIS Poly Screw 6.5 x  35mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS365-035; Known Lot No. WNV101206.
PALLAS M/MAXIMIS 50mm Curved and precontoured Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-050; Known Lot No. WNV120322.
PALLAS M/MAXIMIS 50mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. M058-050; Known Lot No. WNV120322.
PALLAS M/MAXIMIS Poly Screw 6.5 x 55mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P365-055; Lot No. WNV120322.
PALLAS M/MAXIMIS Nut Driver 4.5 Star and Final Driver 4.0. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Driver Shaft (5Hex, Set Screw) 1/4 inch Drive. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Extension Sleeve. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Rod Gauge. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0902-1027. Known Lot No. WNV120723.
PALLAS M/MAXIMIS MIS Screw 6.5 x 45mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 30mm Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Polyaxial Screw and MIS Screw, 5.5 x 40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P755-040; Known Lot No. Unknown, WNV110513, WNV120322, WNV110511.
PALLAS M/MAXIMIS Rod Holder. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Torque Handle. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Reduction B. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 60mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. C060-060; Known Lot No. WNV110516
PALLAS M/MAXIMIS MIS Anti Torque. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 120mm MIS Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. MR58-080; Known Lot No. WNV120424.
Catalog No. P765-035; Known Lot No. WNV110513, WNV120420.
PALLAS M/MAXIMIS 45mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. R060-045; Known Lot No. WNV120322.
PALLAS M/MAXIMIS 5.5 x 05mm Curved Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS C. Probe. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Alignment Driver. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0902-1029. Known Lot No. WNV120328.
PALLAS M/MAXIMIS Persuader. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 7.5 x 55mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P775-055; Known Lot No. WNV120322, WNV120328.
PALLAS M/MAXIMIS MIS Screw 6.5 x 40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Dilator A. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Set Screw Starter. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MIS Screw 5.5 x 50mm and 5.5 x 5.0 mm screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PM755-050; Known Lot No. Unknown, NHS160219, NHS150813, WNV110513, WNV120322.
PALLAS M/MAXIMIS Rod Pusher Angled. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Rod Checker. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. 0902-1026. Known Lot No. WNV100705.
Catalog No. MR58-050; Known Lot No. WNV101020.
PALLAS M/MAXIMIS Dilator B. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS MISS/MIS Poly Screw Driver. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Anti Torque Device. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS 300mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. SR60-300; Known Lot No. NHS160219.
PALLAS M/MAXIMIS 40mm Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. SR60-040; Known Lot No. WNV120328, WNV101022.
PALLAS M/MAXIMIS Poly Screw 6.5 x  40mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. PS365-040; Known Lot No. NHS150813, WNV101213.
PALLAS M/MAXIMIS MIS Screw 6.5 x 45mm and 6.5 x 4.5 screw. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
PALLAS M/MAXIMIS Poly Screw 5.5 x 50mm. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. P355-050; Known Lot No. WNV110513.
PALLAS M/MAXIMIS 500mm Straight Rod. Part of MAXIMIS Pedicle Screw Spinal Fixation System.  Valorem Surgical, Chicago, IL.  Product labeled as Pallas M Spinal Fixation System through 2015.  Intended for posterior pedicle screw fixation of the non-cervical spine.
Catalog No. SR60-500; Known Lot No. NHS160219.
Worldwide Distribution in the states of AK, CT, GA, KY, LA, MA, MD, MN, NC, NM, OH, OK, OR, PA, SC, SD, TN, TX, VA, WA and countries of Hawaii  Australia, Austria, Belgium, Denmark, France, Germany, Hong Kong, India, Indonesia, Ireland, Italy, Japan, Netherlands, New Zealand, Poland, Slovenia, Spain, Sweden, Switzerland, United Kingdom, & Vietnam.
iGUIDE Software 2.2     Product Usage:  Control of accurate patient positioning with assistance of a 30 Tracking System in a radiotherapy environment.
Two or more of the following: Email, Fax, Letter, Press Release, Telephone, Visit
Nationwide Distribution including OH, MA, GA, FL, NC, MS, MO, PA, IN, NJ, TX, KS, LA, NE, WA, AL, UT, AK, CA, IL, MN, MD, NY, and MI.
Hypodermic Needle, 21 G X 1.5 in YALE DISP, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Insufficient documentation to support product sterility claims.
Hypodermic Needle, 27 G X 1.0 in DISP HYPO, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 18 G X 1 in DSP REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 25 G X 5/8, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 20 G X 1.5 in BD YALE DISP RG, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 25 G X 5/8 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 21 G X 1.0, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 22 G X 1.5 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 25 G X 1.0, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 23 G X 1.0, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 5 CC L/L, 100 per box, 20 boxes/case, 2,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 22 G X 1.0 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 10 CC L/L 21 G X 1.0, 100 per box, 10 boxes/case, 1,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 3 CC L/L, 100 per box, 30 boxes/case, 3,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 1 CC L/S, 100 per box, 40 boxes/case, 4,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 60 CC L/S, 50 per box, 8 boxes/case, 400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 10 CC L/L, 100 per box, 12 boxes/case, 1,200/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 25 G X 1.5 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 18 G X 1.5 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 25 G X 1.0 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 10 CC L/L 22 G X 1.5, 100 per box, 10 boxes/case, 1,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 10 CC L/S, 100 per box, 12 boxes/case, 1,200/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 30 G X 0.5 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 5 CC L/L 20 G X 1.5, 100 per box, 15 boxes/case, 1,500/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 5 CC L/L 21 G X 1.0, 100 per box, 15 boxes/case, 1,500/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 27 G X 0.5 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Syringe Bulb, 60 CC 50 pcs per case.    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 30 G X 1.0 in DISP HYPO, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 30 CC L/L, 50 per box, 9 boxes/case, 450/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 23 G X 1.0 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 19 G X 1.0 in BD YALE DISP RG, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 20 CC L/L, 100 per box, 9 boxes/case, 900/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 19 G X 1.5 in BD YALE DISP RG, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 5 CC L/L 22 G X 1.0, 100 per box, 15 boxes/case, 1,500/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 22 G X 1.0, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 22 G X 1.25, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 21 G X 1.5, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 5 CC L/L 22 G X 1.5, 100 per box, 15 boxes/case, 1,500/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 10 CC L/L 22 G X 1.0, 100 per box, 10 boxes/case, 1,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 1 CC TB 27 G X 0.5, 100 per box, 30 boxes/case, 3,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 16G X 1.5in DSP REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 5 CC L/L 21 G X 1.5, 100 per box, 15 boxes/case, 1,500/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 27 G X 1.5 in HYPODERMIC, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 20 G X 1.0, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 10 CC L/L 20 G X 1.0, 100 per box, 10 boxes/case, 1,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 10 CC L/L 20 G X 1.5, 100 per box, 10 boxes/case, 1,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 10 CC L/L 21 G X 1.5, 100 per box, 10 boxes/case, 1,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Hypodermic Needle, 21 G X 1.0 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 20 CC L/S, 100 per box, 9 boxes/case, 900/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 1 CC TB 25 G X 5/8, 100 per box, 30 boxes/case, 3,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 60 CC L/L, 50 per box, 8 boxes/case, 400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 1 CC TB 28 G X 0.5, 100 per box, 30 boxes/case, 3,000/case
Hypodermic Needle, 20 G X 1.0 in REG BEVEL, 100 per box, 50 boxes/case, 5,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 5 CC L/L 20 G X 1.0, 100 per box, 15 boxes/case, 1,500/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 5 CC L/S, 100 per box, 20 boxes/case, 2,000/case  The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 20 G X 1.5, 100 per box, 24 boxes/case, 2,400/case  The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 30 CC L/S, 50 per box, 9 boxes/case, 450/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe and Hypodermic Needle, 3 CC L/L 22 G X 1.5, 100 per box, 24 boxes/case, 2,400/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Piston Syringe, 3 CC L/S, 100 per box, 30 boxes/case, 3,000/case    The products are used to inject fluids into, or withdraw fluids from the body. They are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment or prevention of disease, or to affect the structure or any function of the body.
Worldwide Distribution - US Nationwide in the states of : AR, AZ, CA, GA, IL, IN, KS, MI, MN, MO, NC, NE, NY, OK, PA, SD, TX, & VA. and foreign countries of: Canada, Iran, Mexico, & New Zealand.
Sorin Group Aortic Arch Cannula, 7 mm x 10 in x 3/8 in, Rx Only, Sterile    Product Usage:  The Aortic Arch Cannulae are indicated for use in perfusion of the ascending aorta during cardiopulmonary bypass surgery.
Identification of excess plastic on the tip of the cannula.
Viva ben Femenine Supplement, Dietary Supplement 30 Softgels
Failure to meet with 21 CFR Part 111, CGMPs, assuring identity, purity, and composition of dietary supplements.
Cardiovid Plus, Cardiovascular Active Dietary Supplement, 60 Softgels
ISABEL, Octopus in garlic sauce, Pulpo al ajillo, Net wt - Peso Neto 4 oz(115g)
5,397 Cases (each case has 24 cans (4oz each))
Product label states Octopus and real content is Squid.
ISABEL, Octopus in marinera sauce, Pulpo en salsa marinera, Net wt - Peso Neto 4 oz(115g)
689 Cases (each case has 24 cans (4oz each))
ISABEL, Octopus in vegetable oil, Pulpo en aceite vegetal, Net wt - Peso Neto 4 oz(115g)
5,037 Cases (each case has 24 cans (4oz each))
Fudge Brownie, Net Wt. 9 oz. (0.6 lb.) (255g), with the following brands:  1. Cashwise Gluten Free Bakery,  2.  Coborn's Gluten Free Bakery.
Some Gluten Free Chocolate Chip Bars were incorrectly labeled as Gluten Free Fudge Brownies.  The Gluten Free Chocolate Chip Bars that are incorrectly labeled as Gluten Free Fudge Brownies do contain milk.
Whole Foods Market PB & J Parfait Made With Greek Yogurt; Net Wt 8 OZ.;
Sunneel Health Foods received a call from their customer indicating some of their Vegan Match Cocoa Parfaits were misbranded as PB & J Parfaits. The ingredient statement failed to declare Soy, Almonds and Coconut allergens.
US Nationwide in the states of  CA, CO, FL, IL, IN, LA, MN, MO, NC, NJ, NY, OH, OK, PA, TX, WY, including  PR.
BARD DISPOZ-A-BAG Leg Bag with Flip-Flo Valve with Drain Extension, Extension Tube and Latex Straps, 19 oz.  Packaged as single unit in thermoformed trays.  There are 50 units per case.    Product Usage:  Vinyl leg bag to be used with male external catheters, Foley catheters, or moth other types of urinary catheters, which are intended for use for bladder management including urine drainage, collection and measurement.
Customer complaint of a cut in the side of the leg bag, indicating a potential breach of the product's sterile barrier.
Oxygen Refrigerated Liquid USP UN 1073, Rx only, At Home Medical 200 American Road, Morris Plains, NJ 07950,973-538-0485.
GMP Deviations: The firm does not include an SOP for testing for out of specifications.
WOODSTOCK ORGANIC MATCHA VANILLA OATS, 1.8 oz (50g) cup  UPC: 0-42563-01786-6
Publix Supermarkets, Inc. (Deli Kitchen)
Publix Greenwise Cashew Butter, NET WT 16 oz (1 lb) 454g, Distributed by Publix Super Markets, Inc, Lakeland, FL 33802
The reason for recall was that some jars of Publix GreenWise Cashew Butter may actually contain Almond Butter.
American Red Cross Blood Services - Massachusetts Region
Apheresis Platelets, Leukocytes Reduced, Irradiated
Platelet products, for which required WBC quality control testing was not performed, were distributed.
American National Red Cross (The) - North Central Region
Platelet product, for which required WBC quality control testing was not performed, was distributed.
Glycopyrrolate Injection Solution, 5 mL 1 mg/5 mL (0.2 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222; NDC 52533-028-15
Glycopyrrolate Injection Solution, 2 mL 0.4 mg/2 mL (0.2 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222; NDC 52533-028-16
Ephedrine Sulfate Injection Solution 1 mL,  50 mg/1 mL (50 mg/mL) Single-Dose Syringe,  Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222,  NDC 52533-258-45
Rocuronium Bromide Injection Solution 5 mL, 50 mg/5 mL (10 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-064-15
Glycopyrrolate Injection Solution, 1 mL 0.2 mg/1 mL (0.2 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222; NDC 52533-028-45
Norepinephrine Bitartrate 16 mg Added to 0.9% Sodium Chloride 250 mL (64 mcg per mL) Single-Dose Bag, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222,  NDC 52533-288-18
Adenosine in 0.9% Sodium Chloride 30mL, 90 mg/30 mL (3 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222; NDC 52533-236-03
Norepinephrine Bitartrate 8 mg Added to 0.9% Sodium Chloride 250 mL (32 mcg per mL) Single-Dose Bag, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-217-18
Ephedrine Sulfate in 0.9% Sodium Chloride 5 mL,  50 mg/5 mL (10 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222, NDC 52533-118-15
Diltiazem HCl 125 mg in 5% Dextrose 125 mL Single-Dose- Bag,  Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222; NDC 52533-103-13
Succinylcholine Chloride Injection Solution 5 mL, 100 mg/5 mL (20 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-067-15
Neostigmine Methylsulfate Injection Solution 5 mL, 5 mg/5 mL (1 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-046-15
Phenylephrine HCL in 0.9% Sodium Chloride 10 mL, 1 mg/10 mL (100 mcg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-171-12
Sodium Bicarbonate 8.4% Injection Solution 50 mL, 50 mEq (1mEq/mL) Single-Dose Syringe,  Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222
Norepinephrine Bitartrate 4 mg Added to 0.9% Sodium Chloride 250 mL  (16 mcg per mL) Single-Dose Bag, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-134-18
Ephedrine Sulfate in 0.9% Sodium Chloride 10 mL, 50 mg/10 mL (5 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222, NDC 52533-019-12
Ephedrine Sulfate in 0.9% Sodium Chloride 5 mL, 25 mg/5 mL (5 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd Little Rock, AR 72207 877-666-5222, NDC 52533-019-15
Succinylcholine Chloride Injection Solution 10 mL, 200 mg/10 mL (20 mg/mL) Single-Dose Syringe, Rx Only, Cantrell Drug Company 7321 Cantrell Rd. Little Rock, AR 72207 877-666-5222, NDC 52533-067-12
Nystatin Topical Powder, USP, 100,000 USP units per gram, 15 grams per bottle, Rx only, Vensun, NDC 42543-052-61
Presence of Foreign Substance:  potential presence of plastic particles.
Hydromorphone Hydrochloride Injection, USP, 50 mg/5 mL (10 mg/mL), 5 mL Single Dose Vial.(10 vials per carton NDC 0703-0113-01) and 180 vials per case (NDC 0703-0113-03)  Mfd By: Hospira, Inc., Lake Forest, IL 60045 USA, Mfd For: Teva Parenteral Medicines, Inc., Irvine, CA 92618 USA.
Presence of Particulate Matter: Silicone oil
Hydromorphone Hydrochloride Injection, USP, 500 mg/50 mL (10 mg/mL), 50 mL Single Dose Vial. Mfd For:  Teva Parenteral Medicines, Inc., Irvine, CA 92618 USA. NDC: 0703-0018-01
Hydromorphone Hydrochloride Injection, USP, 500 mg/50 mL (10 mg/mL), 50 mL Single Dose Vial per Carton, 100 vials per case. Hospira, Inc., Lake Forest, IL 60045 USA, NDC: 0409-2634-50
J.J. LAWSON BELGIUM CHOCOLATE ICE CREAM, 3 gallon round container
Product contains undeclared allergen-egg yolk
J.J. LAWSON CHOCOLATE CHOC. CHIP ICE CREAM, 3 gallon round container
Product distributed to OH, IL, PA, MI, VA and CT
Glipzide Extended-Release Tablets (anti-diabetic agent), 5 mg, packaged in 30-unit dose blister pack per carton, Rx only, Mfd for: Watson Laboratories, Inc., Corona, CA 92880, Mfd by: Patheon Pharmaceuticals, Inc., Cincinnati, OH 43215, NDC 0591-0844-15
2,880 cartons (30 unit dose blister pack per carton)
Failed Moisture Limits: out of specification test results for water content obtained during stability testing.