Recalling Firm
Philips Electronics North America Corporation

Initial Firm Notification of Consignee or Public
or more of the following: Email, Fax, Letter, Press Release, Telephone, VisitFLOR
Two or more of the following: Email, Fax, Letter, Press Release, Telephone, Visit

Distribution Pattern
CT, MA, NH, NY, WI, OH, TX, AZ, MO, MN, MI, CA, CO, OR, NM, IN, FL, SC, PA, WA, RI, IL, MD
Distributed in the states of NY, WI, MA, AL, FL, GA, RI, CA, KS, and TX, and the countries of Canada, Australia, Austria, Belgium, France, Germany, Ireland, Italy, Netherlands, New Zealand, Portugal, South Africa, Spain, and Switzerland.
Approximately 335,000 additional cases of the recalled product were shipped by Kraft Heinz to retailers in the U.S. and Puerto Rico, Anguilla, Bahamas, Belize, Bermuda, Cayman Islands Grand Cayman, Netherland Antilles, Saint Kitts and Nevis, South Korea, St. Lucia, and British Virgin Islands.
US nationwide distribution including FL, NJ, OK, and MO.
U.S. distribution to the following; VT, CO, PA. No foreign distribution.
The firm distributed the product to 86 domestic firms. The spread sheet is attached.
Worldwide Distribution. US states of TN and IN and the countries of Australia, Belgium, Egypt, France, Germany, Japan, Republic of Korea, Netherlands, and Saudi Arabia.
Nationwide Distribution-DC and the states of AK, AL, AR, AZ, CA, CO, CT, DE, FL, GA, HI. IA, ID, IL, IN, KS, KY, LA, MA, MD, ME, MI, MN, MO, MS, NC, ND, NE, NH, NJ, NM, NV, NY, OH, OK, OR, PA, SC, SD, TN, TX, UT, VA, VT, WA, WI, and WV.
Worldwide Distribution-US (nationwide) and the countries of Canada, Costa Rica, Libya, Mexico, and Philippines.
RA Food Brokerage, Corp. distributed the affected product to retailers located in Puerto Rico only.
Worldwide Distribution - US (nationwide), Guam, and countries of: Australia, Canada, Croatia, Turkey, New Zealand, Saudi Arabia, Japan, Korea, and Italy.
US, South Africa, Japan and United Arab Emirates.
Domestic: MI, NY, OH, CA, IA, IL, NH, DC, PA, TX, AZ, CO, GA, MT, MS, NV, TN, VA Foreign: None VA/DOD: None
Worldwide Distribution-US (nationwide) including Guam and Puerto Rico and the countries of Algeria, Angola, Argentina, Australia, Bahrain, Belarus, Belgium, Bosnia and Herzegovina, Brazil, Bulgaria, Canada, China, Colombia, Croatia, Czech Republic, Denmark, Dominican Republic, Ecuador, Egypt, Finland, France, Georgia, Germany, Greece, Guadeloupe, Guatemala, Hong Kong, Hungary, Iceland, India, Indonesia, Iran, Iraq, Ireland, Israel, Italy, Japan, Jordan, Korea, Kuwait, Latvia, Lebanon, Libyan Arab Jamahiriya, Lithuania, Macedonia, Malaysia, Mexico, Morocco, Netherlands, New Zealand, Nigeria, Norway, Pakistan, Panama, Philippines, Poland, Portugal, Romania, Russia, Saudi Arabia, Serbia, Singapore, Slovakia, Slovenia, South Africa, Spain, Sudan, Sweden, Switzerland, Syria, Taiwan, Thailand, Tunisia, Turkey, United Kingdom, Uruguay, Venezuela, and Yemen.
Worldwide Distribution -- US, including Washington D.C. Puerto Rico, Virgin Islands, and Guam; and the countries of Afghanistan, Albania, Anguilla, Antigua and Barbuda, Armenia, Australia, Austria, Bahamas, Bahrain, Belgium, Bermuda, Bolivia, Brazil, Bulgaria, Canada, Chile, China, Colombia, Costa Rica, Czech Republic, Denmark, Dominican Rep., Ecuador, Egypt, Ethiopia, Finland, France, Germany, Greece, Guadeloupe, Guatemala, Guyana, Honduras, Hong Kong, Hungary, Iceland, India, Iraq, Ireland, Israel, Italy, Japan, Jordan, Kazakhstan, Kenya, Korea (Republic of), Kuwait, Lebanon, Malaysia, Martinique, Mexico, Netherlands, New Zealand, Nigeria, Norway, Pakistan, Panama, Papua New Guinea, Paraguay, Philippines, Poland, Portugal, Romania, Russian Federation, Saudi Arabia, Singapore, South Africa, Spain, Sweden, Taiwan, Thailand, Trinidad , Turkey, United Arab Emirates, United Kingdom, Uruguay, Venezuela, and Vietnam.
distributed in CA, OR, and TX. Products sold from bulletproof.com to internet customers nationwide and Canada.
Nationwide Distribution to the states of : MN, WI, IL, PA, AL, MO, MT, AZ, CA, OH, FL, SC, MI, TN, CO, NV, IN, NY, MD, AR, KS, NY, IN, IA, TX, OK, NC, WA and DC.
Worldwide Distribution. US Nationwide in the states of NY, CA, KS, NJ, CO, MI, MD, WA, CT, KY, OR, OH, IA, SC, AZ, AK, IN, TN, WI, FL, AL, UT, WY, MA, PA, MN, IL, MI, GA, ND, RI, including Puerto Rico; and the countries of Thailand, France, Germany, Colombia, Hungary, Netherlands, Saudi Arabia, Poland, Spain, and Italy
Distributed in the states of AL, TX, and CA.
US Nationwide Distribution in the states of FL, LA, TX and VA
Worldwide Distribution -- US, Spain, Italy, and Australia.
Worldwide Distribution-US Nationwide, Armenia, Australia, Austria, Belgium, Czech Republic, Denmark, Egypt, France, Germany, Greece, Hungary, India, Ireland, Israel, Italy, Japan, Luxembourg, Malaysia, Netherlands, New Zealand, Norway, P R China, Portugal, Poland, Russia, Singapore, Slovenia, South Africa, Spain, Sweden, South Korea, Switzerland, Thailand, Turkey, UAE, UK, and Brazil.
Worldwide distribution. US Nationwide and countries of AT, AU, BE, CA, CH, CL, CN, CZ, DB, DE, DK, EE, ES, FI, FR, GB, HK, IE, IL, IN, IT, JP, LT, LV, NL, NO, PL, PR, PT, RO, SE, SG, SI, TH, TR, and ZA.
Nationwide distribution to Florida, North Carolina and Pennsylvania.
Worldwide Distribution - US including MA and TX; Foreign distribution to Austria, Azerbaijan, Belgium, British Virgin Islands, Czech Republic, Denmark, Finland, France, Germany, Greece, Italy, Japan, Luxemburg, Morocco, Netherlands, New Caledonia, Norway, Poland, Portugal, Qatar, Romania, San Marino, Spain, Sweden and Switzerland.
Distributed in the states of TX, NY, PA, TN, SC, and OH, and in China.
Worldwide Distribution - US including AL, AK, AR, AK, CA, CO, CT, DC, DE, FL, GA, HI, IA, ID, IL, IN, KS, KY, LA, MA, MD, ME, MN, MI, MO, MS, NC, ND, NE, NH, NJ, NM, NV, NY, OH, OK, OR, PA, PR, SC, SD, TN, TX, UT, VA, WA, WI, WV and Internationally to Canada, Japan, Europe.
Worldwide Distribution. US Nationwide, Venezuela, United Kingdom, Ukraine, Turkey, Thailand, Taiwan, Switzerland, Spain, Saudi Arabia, Russia, Poland, Pakistan, Myanmar, Mexico, Korea, Japan, Italy, Ireland, India, Greece, Germany, France, Finland, Czech Republic, Colombia, China, Canada, and Belgium.

Product Description
ature brand FEP-862S Q STATUS PELLET FORMULA Pelleted Maintenance & Light Exercise Ration 50 lb bag (paper bags, poly-lined, sewn at both ends) 89 b
Primal Formula for Cats; Turkey - 93% Turkey, 7% produce and supplements. 1 oz. nuggets - Keep Frozen: Net wt. 3 lbs. (1.35 kg) Feline Turkey Formula Ingredients: Turkey, Turkey Necks, Turkey Hearts, Turkey Livers, Organic Collard Greens, Organic Squash, Organic Celery, Cranberries, Blueberries, Organic Pumpkin Seeds, Organic Sunflower Seeds, Minerals (Zinc Sulfate, Copper Carbonate, Selenium), Almonds, Organic Apple Cider Vinegar, Sardine Oil, Taurine, Organic Quinoa Sprout Powder, Dried Organic Kelp, Organic Cilantro, Organic Coconut Oil, Cod Liver Oil, Organic Ginger, Natural Vitamin E. Primal Pet Foods Inc.
Philips Healthcare TraumaDiagnost; Codes: 72001, 720011, 720013, 720020; A radiological examination unit for emergency work and routine skeletal diagnosis.
START A300 PAC, 2000# tote, Manufactured by Provimi North America, Inc., Brookville, Ohio 45309
Kraft Singles White American singles packaged in a blue box with white lettering.
Kraft Singles Deli Deluxe American Slices packaged in a blue box with white lettering.
Kraft Singles American packaged in a blue box with white lettering.
CORDEL GF5 DDG 20 C-M, Swine Grower/Finisher, Complete to Increase the Digestibility of Phytin-Bound Phosphorus in Swine Diets
DQ Goat Grower 15% RU-27 Medicated - A Feed for Goats, Monensin Sodium 27 g/ton, packaged as bulk in 1000 lb. plastic totes. The responsible firm is MFA Incorporated, Columbia, MO 65201.
OC Raw Dog Chicken, Fish & Produce Raw Frozen Canine Formulation 3lb Meat Rox UPC Code: 022099069171 4lb Sliders UPC Code: 095225852640 6.5lb Doggie Dozen Patty Bags UPC Code: 022099069225
Morphine Sulfate, 2 mg per mL (60 mg per 30 mL) 30 mL total volume in a 35 mL Monoject Barrel Syringes in Sodium Chloride 0.9%, Rx only, Contains Sulfites, Compounded Drug, Not for Resale, Hospital/Office Use only, IV use only, NDC 61553-401-44.
Inclusion bar-Cocao Nibs, caramel almonds, and sea salt.
Meltaway bar- French Vanilla & salted Almond.
Inclusion chocolate bar 66% Cacao topped with pistachios, cranberries & Sea Salt in a white and green wrapping with white and gray lettering.
Inclusion bar-macadamia nuts, raspberries, and kaffir lime
Inclusion bar-strawberries, rose petals, and pink peppercorns.
Meltaways- 3 piece peppermint meltaways.
Meltaways- 3 piece Blood Orange meltaways.
Inclusion bar-expresso, cherries, and hazelnuts.
Inclusion bar-candied ginger, candied orange peel, and coriander.
Inclusion bar-cherries, cocoa nibs, cayenne.
Mendiant- Pistachio, Rose Petal & Candied Ginger.
Meltaway bar- Blood Orange & Vanilla bean.
Mendiant Orange apricot & carmelized hazelnut packaged in a clear plastic bag with brown label with white lettering.
Meltaway bar- Sweet basil & Cherry blossom.
Inclusion bar-scorpion pepper, bourbon, smoked sea salt, and caramel pecans.
Meltaways- 3 piece Vanilla Sea Salt meltaways.
Inclusion bar-strawberries, mangoes, and chiles.
Meltaways- 3 piece Tahitian Vanilla Lemon Meltaways.
Philips Allura Xper FD20 Ceiling version only; 722028; imaging applications.
Allura Xper FD OR Table, Ceiling version only; 722035; imaging applications.
Arctic Sun 5000 and Arctic Sun 2000; Intended for monitoring and controlling patient temperatures.
VS300 radiographic Cassette Holder or Wall Stand. For use in radiology department.
VS200 radiographic Cassette Holder or Wall Stand. For use in radiology department.
WATCHAREE'S Thai Massaman curry sauce (UPC 094368999984), product is in a glass jar, NET WT 11.5 OZ (326 g)
WATCHAREE'S thai peanut sauce (UPC 094368999977), product is in a glass jar, NET WT 11.75 OZ (333 g)
Watcharee's Thai Green Curry Sauce (UPC 094368999991), product is in a glass jar, NET WT 11.5 OZ (326 g)
SyrSpend SF, Suspending Base, a convenient, sweetened, sugar-free syrup-suspending vehicle specially formulated to assist in extemporaneous preparation of oral liquid dose forms, Rx Only, Shake Well, 4 L bottle, Manufactured by Fagron Inc., St. Paul, MN 55120, NDC 51552-1079-9.
SyrSpend SF, Suspending Base, Grape Flavored, a convenient, sweetened, sugar-free syrup-suspending vehicle specially formulated to assist in extemporaneous preparation of oral liquid dose forms, Rx Only, Shake Well, 4 L bottle, Manufactured by Fagron Inc., St. Paul, MN 55120, NDC 51552-1167-9.
SyrSpend SF, a convenient, sweetened, sugar-free syrup-suspending vehicle specially formulated to assist in extemporaneous preparation of oral liquid dose forms, Rx Only, Shake Well, 500 mL bottle, Manufactured by Fagron Inc., St. Paul, MN 55120, NDC 51552-1079-5.
Moog Curlin Infusion, Ambulatory Infusion Pump The Curlin Ambulatory Infusion System includes Curlin Ambulatory multi-therapy and PainSmart pumps. Curlin 4000, catalog number 340; Curlin 6000, 6000 CMS, and PainSmart IOD, catalog number 360; and Curlin Pump Calibration Set, catalog number 340-4038.
Mizuho OSI Trios, Table Base; Voltage: 100-240V ~ 50/60 Hz Amps: 5,0-5.0A; Model 7803 and 7803AP Orthopaedic: Operating room tablebase - an electronically powered mobile operating table is created, designed for temporary (<24 hours) support and positioning of a patient in a prone, supine, or lateral position depending on the specific table top utilized.
Haloperidol Oral Solution, USP (Concentrate) 2 mg/mL 15 mL amber bottle, Rx Only, Manufactured by Silarx Pharmaceuticals Inc. Carmel, NY 10512, Packaged by Safecor Health, Woburn, MA 01801, NDC 54838-0501-40.
Zzang! Holiday 4 pack Candy Bars, 10.5 oz., packaged in preprinted cardboard boxes
Zzang! Cashew Cow Candy Bar,2 oz., packaged in reprinted cardboard boxes, UPC 8 44397 00015 7
Chocolate Marshmallows, 8 oz., packaged in plastic bags, UPC 8 44397 00077 5
Zzang! Wowza Candy Bar,3 oz., packaged in reprinted cardboard boxes, UPC 8 44397 00072 0
Zzang! Original Candy Bar, 2.5 oz., packaged in preprinted cardboard boxes, UPC 8 44397 00014 0
Bearded Brothers BOLD The Greek Net Wt. 1.5 oz/43 g UPC: 8-50430-00414-2.
Bearded Brothers BOLD Coconut Curry Net Wt. 1.5 oz/43 g UPC: 8-50430-00513-5.
Mavig Monitor Suspension System, Model GD60, contained in the following Fluoroscopic imaging systems: Advantx LCV+, Advantx LCN+, Advantx LCLP+, Advantx LCA, INNOVA 2000, INNOVA 4100, INNOVA 3100, INNOVA 3131IQ, INNOVA 2121IQ, INNOVA 2100IQ, INNOVA 3100IQ, and INNOVA 4100IQ
Midazolam HCl Syrup, 10 mg/5 mL, 5 mL unit dose cups, packaged in 2 shelf boxes, one shelf box containing 72 cups and one shelf box containing 60 cups, Cherry Brandy Flavor, Rx only, MFG: Roxane; PKG BY Safecor, Woburn, MA; Barcode 0054356605.
GE Healthcare, Optima XR220amx, Mobile Digital Ready Radiographic System Designed to generate diagnostic radiographic images.
GE Healthcare, Brivo XR285amx Mobile Digital Ready Radiographic System Designed to generate diagnostic radiographic images.
GE Healthcare, Optima XR200amx, Mobile Digital Ready Radiographic System Designed to generate diagnostic radiographic images.
FATwater bottle, 16.9 fl. oz., Berry flavor, UPC 6 09015 50477 2
FATwater bottle, 16.9 fl. oz., Tangerine flavor, UPC 6 41871 93693 9
FATwater bottle, 16.9 fl. oz., Lemon flavor, UPC 6 09015 50475 8
IMPLANT,TM, MTX 3.7 mm x 11.5 mm diameter 3.5 mm Catalog #TMMB11 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 3.7 mm x 11.5 mm diameter 3.5 mm Catalog #TMTB11 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 4.1 mm x 10 mm diameter 3.5 mm Catalog #TMT4B10 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 4.1 mm x 11.5 mm diameter 3.5 mm Catalog #TMT4B11 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 3.7 mm x 10 mm diameter 3.5 mm Catalog #TMTB10 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 4.1 mm x 13 mm diameter 3.5 mm Catalog #TMT4B13 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX 3.7 mm x 13 mm diameter 3.5 mm Catalog #TMMB13 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX 4.1 mm x 13 mm diameter 3.5 mm Catalog #TMM4B13 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX 4.1 mm x 10 mm diameter 3.5 mm Catalog #TMM4B10 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 3.7 mm x 16 mm diameter 3.5 mm Catalog #TMTB16 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX 4.1 mm x 11.5 mm diameter 3.5 mm Catalog #TMM4B11 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX 3.7 mm x 10 mm diameter 3.5 mm Catalog #TMMB10 Product Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
IMPLANT,TM, MTX FULL 3.7 mm x 13 mm diameter 3.5 mm Catalog #TMTB13 Product Usage: Zimmer Dental implant systems are designed for use in the maxilla or mandible for immediate loading or for loading after a conventional healing period. Implants may be used to replace one or more missing teeth. Immediate loading is indicated when there is good primary stability and an appropriate occlusal load.
Pramipexole Dihydrochloride Tablets 0.125 mg, 90 Ct Bottles, Rx Only. Manufactured for: Macleods Pharma USA, Inc., Plainsboro, NJ 08536, Manufactured by: Macleods Pharmaceuticals Ltd., Baddi, Himachal Pradesh, INDIA. NDC: 33342-031-10.
Bard Distaflo Vascular Bypass Graft; Product Code: DF8006SC; 6 mm x 80 cm. Intended for bypass or reconstruction of peripheral arterial blood vessels.
Bard Distaflo Vascular Bypass Graft; Product Code: DF6006SC; 8 mm x 60 cm. Intended for bypass or reconstruction of peripheral arterial blood vessels.
Candesartan Cilexetil Tablets, 16 mg, a) 30 count (NDC 00781-5938-31) and b) 90 count (NDC 00781-5938-92) bottles, Rx Only, Manufactured by Mylan Laboratories Limited Hyderabad 500 034, India for Sandoz Inc. Princeton, NJ 08540
Signature Kitchens Crushed Wheat Hot Dog; 8 buns; NET WT 12 oz 340 g UPC 0 21130 18496 5; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Signature Kitchens Enriched Hamburger Buns, Sesame; 8 buns; NET WT 12 oz 340 g UPC 0 21130 18467 5; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Pantry Essentials Hot Dog Buns; 8 buns; NET WT 12 oz 340 g UPC 0 21130 27084 2; Distributed for: Lucerne Foods Inc. Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Signature Kitchens Enriched Hot Dog; 8 buns; NET WT 12 oz 340 g UPC 0 21130 18501 6; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Signature Kitchens 100% Whole Wheat Hot Dog Buns; 8 buns; NET WT 16 oz 454 g UPC 0 21130 15456 2; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Pantry Essentials Hamburger Buns; 8 buns; NET WT 12 oz 340 g UPC 0 21130 27085 9; Distributed for: Lucerne Foods Inc. Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Coast to Coast San Francisco Style Sourdough 6 Hoagie Rolls; 6 Hoagie Rolls; NET WT 16.5 oz 468 g UPC 0 21130 18091 2; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Safeway Kitchens Enriched Hamburger Buns; 8 buns; NET WT 17 oz (1 lb 1 oz) 482 g UPC 0 21130 18456 9; Distributed by:Safeway Inc. Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
Old Fashioned Wheat Diner Rolls 8 Count Pack size 20 oz. UPC 2 24247 00000
Old Fashioned White Diner Rolls 8 Count Pack size 20 oz. UPC 2 24246 00000
Open Nature Whole Grain White Hot Dog Buns; 8 buns; NET WT 12 oz (340 g) UPC 0 79893 11057 8; Distributed by: Lucerne Foods, inc. Pleasanton, CA 94566 Contains: Wheat, milk May Contain egg, milk.
Coast to Coast Traditional Hawaiian Sweet 12 Dinner Rolls; 12 Dinner Rolls; NET WT 14 oz 397 g UPC 0 21130 18580 1; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat. milk May Contain eggs
Open Nature Seeded Sandwich Rolls; 8 buns; NET WT 21 oz (1 lb 5 oz) 595 g UPC 0 79893 11439 2; Distributed by: Lucerne Foods, inc. Pleasanton, CA 94566 Contains: Wheat May Contain egg, milk.
Coast to Coast Parisian Style French 6 Hoagie Rolls; 6 Hoagie Rolls; NET WT 16.5 oz 468 g UPC 0 21130 18090 5; Distributed by Living Better Brands, LLC, Pleasanton, CA 94566 Contains: Wheat May Contain eggs, milk.
PharmaCaribe PulmoSal 7% Sodium chloride, Inhalation Solution, 7% Sterile, Preservative -free, Non-pyrogenic, pH 7.4 For Inhalation Only Rx Product Usage: The contents of these vials are used in conjunction with a nebulizer for the induction of sputum production where sputum production is indicated.
Firebird Spinal Fixation System Torque Limiting Handle (PN 52-1512) Intended for posterior, non-cervical pedicle, and non-pedicle fixation (T1-S2/llium).
SIGMA HP MBT Non-Keel Punch Knee Instrument. Designed to be used as an option in stabilizing the tibial trial during trial reduction.
INFX-8000V Bi-Plane X-Ray Interventional System X-ray systems
Smith & Nephew BEAVER Blade 4.0mm Part Number: 72203307 Orthopedic surgical instrument used in hip arthroscopy.
Reuseable rotary blade/abrader, product number 899751504. The oval burr laterally hooded is an attachment or accessory to the air-powered device intended for use during surgical procedures to cut hard tissue or bone and soft tissue. General hospital and plastic surgery use.
Reuseable rotary blade/abrader, product number 899751514. The oval burr frontally hooded is an attachment or accessory to the air-powered device intended for use during surgical procedures to cut hard tissue or bone and soft tissue. General hospital and plastic surgery use.
Reuseable rotary blade/abrader, product number 899751305. The round burr is an attachment or accessory to the air-powered device intended for use during surgical procedures to cut hard tissue or bone and soft tissue. General hospital and plastic surgery use.
Patient transport Shuttle with Trendelenburg quick adjustment function (Jupiter, Mars, Titan, TruSystem and Saturn operating table systems). The shuttle is intended for the following applications: 1. Transfer, transport and storage of exchangeable table tops in the Trumpf Medical TruSystem 7500, JUPITER, and SATURN SM operating table systems 2. Patient transport to an operating table top (without a column) from the patient transfer system to the operating theatre or from the operating theatre to the patient transfer system 3. Transporting the TruSystem 7500 SM, JUPITER SM or SATURN SM operating table column with table top (without a patient) within the operating theatre.
T-piece Nebulizer with in-line connectors, Part number 8915-7-50. Single patient use, non-sterile prescription device designed for use in a hospital or home care environment.
Filling alveoli after tooth extraction. This product should not be used in non-peridontal mandibular applications.
CyberKnife Robotic; Catalog/part number 032000 and 033000 Cosmetic cover package. Product Usage: The CyberKnife Robotic Radiosurgery System is indicated for treatment planning and image guided stereotactic Radiosurgery and precision radiotherapy for lesions, tumors, and conditions anywhere in the body when radiation treatment is indicated.

Product Quantity
52 cases (8 x 3 lb. bags per case) (1248 lbs.
399 distributed, 262 are subject to this event
12, 3 lb. cartons, 8, 4 lb. cartons, 12, 1 lb. cartons.
6, 3 lb. cartons. 36, 1 lb. cartons. 12, 3 lb. cartons. 8, 4 lb. cartons.
14,433 Original Candy Bars; 3,242 Cashew Candy Bars ; 1,499 Wowza Candy Bars

Reason for Recall
Samples collected by the Colorado Department of Agriculture and tested in FDA laboratory tested positive for Salmonella
One lot of Feline Turkey Raw Frozen Formula 3 pound bag tested FDA was found to be low in Thiamine.
Product tested positive for Salmonella spp. Poly A
Product did not test positive for Salmonella, but is being voluntarily recalled out of an abundance of caution because it was manufactured in the same manufacturing facility on the same day as the product that tested positive for Salmonella spp. Poly A.
If arm is not fastened securely it could fall down.
Elevated levels of vitamin A can lead to elevated levels of morbidity and mortality in swine.
Kraft Singes wrapped American slices due to the possibility that a thin strip of the individual packaging film may remain adhered to the slice after the wrapper has been removed. If the film sticks to the slice and is not removed, it could potentially cause a choking hazard.
The treatment table positioning belt can slip gear positions, resulting in deviation of patient setup and treatment positions from the time of slippage until the discovery of slippage and the halting of treatment.
Product contains monensin and excess copper.
OC Raw Dog, LLC is recalling OC Raw Dog Chicken, Fish & Produce Raw Frozen Canine Formulation due to potential Salmonella contamination.
Sub-potent Drug; firm's analysis revealed subpotent result for morphine sulfate assay.
Undeclared Milk on the labels of various dark chocolate bars. FDA samples came back positive for milk.
Normally, movement of the C-arc is initially stopped by the motor (holding torque) then by the electromagnetic brake, leading to a full stop of the C-arc. Because of improper adjustment and tolerances, when the C-arc is initially stopped in an unbalanced position, the air gap of the brake can become too large, resulting in continued C-arc movement (C-arc is looking for balanced position).
Improper targeted temperature therapy was delivered to patients because the Temperature Simulator Key was connected to the patient temperature cable instead of the patient temperature probe.
The wall stand cassette holder can suddenly fall to the bottom of its travel, near the ground and contact patient.
Sauces may not meet quality standards due to the presence of mold.
Microbial Contamination of Non-Sterile Products: Syrspend SF and Syrspend SF Grape Flavor are being recalled due to the presence of yeast.
Product in Import Alert (99-38) inadvertently released for distribution.
Moog is recalling Curlin 4000, 6000, and PainSmart Infusion Pumps manufactured or serviced by MMDG between March 18, 2015 and November 6, 2015 because the master pumps used to calibrate them were out of tolerance and may cause an over delivery of medications and solutions at a rate of 1.2 - 6.8% faster than programmed.
Table Bases may have been assembled incorrectly, missing a retaining ring from the head end assembly. Over time, the parts could go out of alignment, making clutch engagement or disengagement difficult. If the clutch cannot disengage, the user would be unable to rotate the patient from supine to prone.
Defective Container: Confirmed customer complaints of leaking bottles.
Product recalled due to undeclared soy; and undeclared allergen milk (cream and butter declared).
Product recalled due to undeclared sub-ingredient soy.
Product recalled due to undeclared soy; and undeclared milk (cream declared).
Product recalled due to undeclared soy and milk (butter declared).
GE Healthcare has recently become aware of a reported incident in which a Mavig suspension arm with monitor assembly fell to the floor.
Presence of Foreign Substance: customer complaint that one unit dose cup contained a small piece of cardboard contaminant.
Potential safety latch mechanism failure. The safety latch, within the column that supports the horizontal arm, is a backup in the unlikely event the primary safety cable breaks. A fall of the arm assembly could result in bodily harm to a person.
FATwater bottles, Berry flavor, are recalled due to microbial growth detection. No E.coli or other pathogens were detected.
FATwater bottles, Tangerine flavor, are recalled due to microbial growth detection. No E.coli or other pathogens were detected.
FATwater bottles, Lemon flavor, are recalled due to microbial growth detection. No E.coli or other pathogens were detected.
Potential incorrect Standardized Uptake Values (SUV) measurements in RadSuite.
Information specific to the 3.7 mmD and 4.1 mmD Trabecular Metal Zimmer Dental Implants was omitted from the latest revision of the applicable IFU in the Indications, Contraindications, and Precautions sections.
Presence of Foreign Tablets/Capsules: Presence of a comingled Carbimazole 5 mg tablet.
Size mislabeling; the graft is a 6 mm diameter graft when the actual packaged product is an 8 mm graft.
Size mislabeling; the graft is labeled as a 6 mm diameter graft when the actual packaged product is an 8 mm graft.
Failed Impurities/Degradation Specifications; 9 month stability timepoint
Cloth material that is part of the production equipment may be in the product.
Vials were labeled as USP 7% Hypertonic saline instead of PulmoSal 7% (pH+)
There is a possibility that the device may deliver a torque value less than the required setting after initial sterilization. In worst case, the unexpected drop in torque value may lead to the spinal construct loosening and requiring corrective surgical intervention.
Apheresis products, in which leuko reduced units did not meet WBC count specification, were distributed.
Increased punch height, causing a 0.063 in (1.6 mm) gap between the trial and the instrument. This may cause the surgeon to select an insert that is too thin. Surgical delay may result if the punch dislodges from the inserter and needs to be removed from the joint space. Insert Dislocation or Spin Out, or Poor Joint Mechanics may result, if not noted during surgery. May require revision surgery.
Blade exhibiting fracture during hip arthroscopy procedures
Weak weld in the reusable rotary blade/abrader (burr) may result in inner part breaks at the distal or proximal end making the product dysfunctional.
The product is labeled with an expiration date that is past its shelf life.
Complaints were received of the table top on the transporter (shuttle) tilting down quickly in a maximum Trendelenburg position.
Reports that the T-piece does not fit on the nebulizer.
Cytotoxicity tests were not passed at 36 months. Use of Endobon-Xenograft Granules past 18 months of its specified shelf life may result in patient illness/injury.
The gun box mounting bracket may fail to support the weight of the gun box when in vertical (inverted) position. If this failure occurs the gun box may become loose and could come into contact with a patient.

Code Info
Number 14-280 Manufacture Date: October 7
All TraumaDiagnost products that were delivered before the week of July 2, 2007.
Best When Used By date between 12 DEC 15 and 13 FEB 16, Best When Used By date of 12 JAN 16, Best When Used By date of 11 JAN 16
Best When Used By date between 21 JAN 16 and 02 MAR 16
Best When Used By date between 13 DEC 15 and 18 FEB 16, Best When Used By date between 09 DEC 15 and 19 FEB 16, Best When Used By date between 24 JAN 16 and 6 FEB 16.
Manufacturing Lot or Serial Number: S250-0004
Sku: CB1022 expiration date range of 10/21/2015 to 4/15/2016
Sku: MDPRC Expiration date range: 10/21/2015 - 1/15/2016
Sku: MDOACH Expiration date range: 10/21/2015 - 4/15/2016
All lots/serial numbers sold prior to August 2014.
All manufactured or serviced by MMDG between March 18, 2015 and November 6, 2015. Curlin 4000, catalog number 340; Curlin 6000, 6000 CMS, and PainSmart IOD, catalog number 360; and Curlin Pump Calibration Set, catalog number 340-4038.
Model numbers 7803, 7803AP. Serial numbers 228-261.
Lot and Batch Nos are the sell-by dates 12/17/2015 03/16/2016, Product/Catalog No. 100
Lot and Batch Nos are sell-by dates 12/17/2015 02/16/2016, Product/Catalog No. 120
Lot and Batch No is sell-by date 12/24/15 (only 1 batch).
Lot and Batch Nos are sell-by dates 12/17/2015 02/16/2016, Product/Catalog No. 160
LOT number: B151005-4U. EXPIRATION DATE: 10/5/2016.
Product Code: DF8006SC Lot Number: VTZE1783
Product Code: DF6006SC Lot Number: VTZE1782
Lot # CIY Exp.Date August 2017 UPC Number: 030915339988 (on the box of 60 vials) UPC Number: 10030915339985 (on the shipping case of 12 boxes)
Cat. No. GTIN Lot Description 950502016 10603295227366 ABB92402 SIGMA HP MBT Cemented Punch size 1-1.5 950502017 10603295227373 ABB84491 SIGMA HP MBT Cemented Punch size SZ 2-3 950502018 10603295227380 ABB92358 SIGMA HP MBT Cemented Punch size SZ 4-7 950502020 10603295227403 ABC33125 SIGMA HP MBT Noncemented Punch size 2-3 950502020 10603295227403 ABB81733 SIGMA HP MBT Noncemented Punch size SZ 2-3 950502020 10603295227403 ABB81732 SIGMA HP MBT Noncemented Punch size SZ 2-3 950502021 10603295227410 ABB39952 SIGMA HP MBT Noncemented Punch size SZ 4-7 950502021 10603295227410 ABC33123 SIGMA HP MBT Noncemented Punch size SZ 4-7
Product: Shuttle 2.7, Material Number: 1254362, Serial Number Range: 100801175 101349420; Product: Shuttle 2.7 360, Material Number: 1459862, Serial Number Range: 100843669-101212847; Product: Shuttle 3.7, Material Number: 1254363, Serial Number Range: 100180383-101290987 and Product: Shuttle 3.7 360, Material Number: 1459863, Serial Number Range: 100787418-101334652.